I have reverse correction turned off for Omnipod 5 but anytime I was under target the dang thing would always do a reverse correction for my meal bolus. Most of the time I would catch it and manually change it but sometimes if I was distracted I wouldn’t and then of course go much higher than I like. If I am lower than the 110 target OP5 has been suspending insulin for quite awhile leading up to a meal so I think the full meal dose and a pre-bolus is actually more important the lower your BG.
Well turns out that if you manually type in your BG iunder Glucose instead of the option to USE CGM then no reverse correction!
This only took me a year plus to figure out so I’m not feeling so smart right now. Lol